Morehouse Calibration Software Instruction Manual The Morehouse calibration software manual discusses software installation, troubleshooting, and operation of Morehouse free-to-download software.
The manual discusses data acquisition using the coefficients from the calibration report. This is the most accurate method to obtain force measurement readings.
The mV/V readings are converted to force units, and as such, the "as found" data is always the same as the "as left," meaning adjustments are done using the coefficients from the calibration report.
Morehouse Verification Sheet for Legacy Software: This is a one-sheet verification for the Morehouse Calibrator legacy software. The verification verifies the software converts the mV/V values properly.
The end-user must still verify they have entered the coefficients correctly. The sheet provides the verification documentation necessary for audits.
Morehouse Calibration Software
Morehouse calibration software allows readings from any number of Morehouse indicators that have a communication output (USB or serial). If multiple indicators are used, their values can be added together to show a total value.
The software displays values by converting the readings from the indicator(s) using the calibration coefficients (typically from ASTM E74 or ISO 376 calibrations). Additionally, the software can do the unit conversion (between force units and mass units), and mass conversion, and adjust the resolution at the values shown.
Morehouse force software verification Excel sheet. The Excel sheet verifies the force software from Morehouse. The sheet includes formulas and screenshots to prove the Morehouse force calibration does what it should. Download our force verification sheet for force verification.
Software for reading two indicators at the same time
V1.3.5: Force calibration software for use with Morehouse 4215, HADI, or DSC-USB Indicators. Read two indicators at the same time.
The software uses coefficients from ASTM E74 calibration and displays engineering units as well as capturing data with a remote control device and pausing readings etc..,
Reporting section helps with ASTM E4 field calibrations. Must have Morehouse 4215, HADI, or DSC-USB indicator to use this software.
The software is designed to work with Morehouse indicators, Dual Channel 4215, 4215 HS, 4215, 4215 plus, HADI, and the DSC USB.
The torque installer software is the universal version for torque calibration which allows for lbf ft and N m coefficients to be entered. PEAK HOLD and new windows 8/10 drivers on this version. Inverse B0 coefficients.