Load Cell Alignment Plugs Proper Use
Morehouse load cell alignment plugs are used to help center the load cell in calibrating machines. They are used in combination with the other adapters below. More information on load cell adapters and the mounting accessories can be found here.
|Morehouse Reference Standard Mounting Kits for Universal Calibrating Machines|
Proper way to thread a load cell alignment plug. Thread is past flush and into the cell.
When using your alignment plugs that thread into the bottom of your load cells, make sure they are threaded flush to the bottom of the cell. Once they are flush, thread the adapter an extra turn into the cell. You want to make sure that none of the threads are exposed below the base of the cell. If there is a thread or more exposed, the load will be generated through the internal threads of the cell and not it's base. This will result in an additional calibration error of about 0.012 %. It will often result in damage to the alignment plug.
Improper way to thread a load cell alignment plug. The load cell alignment plug is not threaded past the base of the cell.
Load Cell Alignment Plugs Proper Use - Conclusion
Everything we do, we believe in changing how people think about force and torque calibration. Morehouse believes in thinking differently about force and torque calibration and equipment. We challenge the "just calibrate it" mentality by educating our customers on what matters, and what causes significant errors, and focusing on reducing them. Morehouse makes our products simple to use and user-friendly. And we happen to make great force equipment and provide unparalleled calibration services.
Wanna do business with a company that focuses on what matters most? Email us at firstname.lastname@example.org.
Written by Henry Zumbrun with help from the calibration lab technicians, who have received numerous load cells with damaged alignment plugs.