Morehouse Load Cell Aligned vs Misaligned Loading Test
S Beam load cell Aligned vs Misaligned Loading
Force Measurement Error - Loading load cell against the base versus loading through the threads
Cable Length Comparison why 6 wire cable is better
Load Cell Error Demonstration With a Shear Web Load Cell
NASA Webinar: Measurement Traceability and Errors Related to Force Measurement
Proper Adapters to Reduce Force Measurement Error 50 min
May 1, 2018 11:00 AM - 11:50 AM Eastern Time Zone
The focus of this webinar is to discuss the common measurement errors and how to reduce or eliminate them using the proper adapters. These common errors can range from a magnitude of 0.01 % to as high as 0.5 % on the force measuring equipment that is supposed to have uncertainties or accuracies from 0.005 % to 0.1 %. In almost all cases the errors from not using the proper adapters put the claimed uncertainty or accuracy of the force measuring instruments well above what is claimed.
The training was excellent, and the hospitality of everyone in your organization was splendid, and it was an overall pleasure.
Navy Calibration Technician
The biggest take away from my time at Morehouse was your passion for accuracy in everything you do and your desire to improve measurement accuracy regardless of how profitable it may be for you personally. That is impressive.
Field Calibration Technician
I thoroughly enjoyed your Force Class. I also learned something in the process. I appreciate your delivery and efforts to engage everyone attending. The interactive element and your candor made the class the most beneficial to me.